|doseBadge Noise Dosimeter
Measuring Workers Exposure to Noise
Noise Measurement Regulations
The doseBadge meets the requirements of both the American (OSHA) and European (ISO) measurement standards. As it is not actually possible to meet both at the same time, the doseBadge can be configured by the user to either OSHA or ISO settings.
The doseBadge also meets the requirements of the US DOE, ACGIH, AICHE and MSHA regulations and guidelines.
The selection of which standard to measure by is done using the simple menu in the Reader. The doseBadges are shipped pre-configured to the relevant standard, so making the changes should not normally be necessary.
The most common settings are:
Noise Dosimeter Standards
The doseBadge system meets the following standards:
ANSI S1.25:1991 Personal Noise Dosimeters
IEC 61252:1993 Personal Sound Exposure Meters
To satisfy the OSHA measurement standards, the doseBadge measures the following parameters: LAVG, TWA and% Dose. The exchange rate is set to Q=5.
To satisfy the ISO measurement standards, the dosebadge measures the following parameters: LAeq, LEPd, SEL, % Dose, Exposure Pa²h, Estimated % Dose, Estimated Pa²h.
It can easily be set to measure both OSHA style 5dB measurements and ISO 3dB measurements at the same time as it is a dual-channel dosimeter.
|UK 0845 230 2434 :: Int +44 1723 firstname.lastname@example.org|